E6000 Electrical handheld 3 phase power quality analyzer
Troubleshoot real-time: Analyze the trends using the cursors and zoom tools-even while background recording continues. Automatic Transient Mode: Capture 200 kHz waveform data on all phases simultaneously up to 6 kV Measure all three phases and neutral: With included four current probes System-Monitor: Up to ten power quality parameters on one screen according to EN50160 power quality standard View graphs and generate reports: With included analysis software Logger function: Configure for any test condition with memory for over 400 parameters at user defined intervals. Mains signaling: Measure interference from ripple control signals at specific frequencies Inrush mode: For troubleshooting nuisance circuit breaker tripping. AutoTrend: Every measurement you see is always automatically recorded, without any setup. Fully Class-A compliant: Conduct tests according to the stringent international IEC 61000-4-30 Class-A standard. Battery Life: Seven hours operating time per charge on NiMH battery pack High safety rating: 600 V CAT IV/1000 V CAT III rated for use at the service entrance
• Comparing to online power quality analyzer, E6000 handheld type with small size, light weight, easy for the engineer to carry and carry out to survey work whenever and wherever.
• Easy to wiring
• It can operate directly to the line under test under the circumstances that the Power Grid without interrupt.
• Multi-range current switching
• based on actual demand, the different range of current clamp can be replaced to improve measurement accuracy.
Frontline troubleshooting - quickly diagnose problems on-screen to get your operation back online Predictive maintenance - detect and prevent power quality issues before they cause downtime Quality of service compliance - validate incoming power quality at the service entrance Long-term analysis - uncover hard-to-find or intermittent issues Load studies - verify electrical system capacity before adding loads Energy assessments - quantify energy consumption before and after improvements to justify energy saving devices.